Arduino Energy meter
I belong to a village of Odisha, India where frequent power cut is very common. It hampers the life of every one. During my childhood days continuing studies after dusk was a real challenge. Due to this problem I designed a solar system for my home on a experimental basis. I used a solar panel of 10 Watt ,6V for lighting few bright LEDs. After facing lot of hardships the project was successful. Then I decided to monitor the voltage, current, power & energy involved in the system. This brought the idea of designing an ENERGY METER.I used ARDUINO as the heart of this project because it is very easy to write code in its IDE and there are huge numbers of open source library available in the internet which can be used according to the requirement.I have experimented the project for very small rated (10Watt) solar system but this can be easily modified to use for higher rating system.
Energy monitoring by
2. via internet (Xively upload)
3. Data logging in a SD card
You can also see my other instructables on
Step 1: Parts Required :
1. ARDUINO UNO (Rev-3)
2. ARDUINO ETHERNET SHIELD
4. ACS 712 CURRENT SENSOR
4. RESISTORS (10k ,330ohm)
5. POTENTIOMETER (10K)
6. JUMPER WIRES
7. ETHERNET CABLE (CAT-5E)
8. BREAD BOARD
Step 2: Power and Energy
Power is product of voltage (volt) and current (Amp)
Unit of power is Watt or KW
Energy is product of power (watt) and time (Hour)
Unit of Energy is Watt Hour or Kilowatt Hour (kWh)
From the above formula it is clear that to measure Energy we need three parameters
Step 3: Voltage Measurement
Voltage is measured by the help of a voltage divider circuit.As the ARDUINO analog pin input voltage is
restricted to 5V I designed the voltage divider in such a way that the output voltage from it should be less than 5V.My battery used for storing the power from the solar panel is rated 6v, 5.5Ah.So I have to step down this 6.5v to a voltage lower than 5V.
I used R1=10k and R2 =10K. The value of R1 and R2 can be lower one but the problem is that when resistance is low higher current flow through it as a result large amount of power (P = I^2R) dissipated in the form of heat. So different resistance value can be chosen but care should be taken to minimize the power loss across the resistance.
Vbat=6.5 when fully charged
R1=10k and R2=10k
Vout=10/(10+10)*6.5=3.25v which is lower than 5v and suitable for ARDUINO analog pin
I have shown 9 Volt battery in bared board circuit is just for example to connect the wires.But the actual battery I used is a 6 Volt, 5.5Ah lead acid battery.
When battery is fully charged (6.5v) we will get a Vout=3.25v and lower value for other lower battery voltage.
AEDUINO ADC convert Analog signal to corresponding digital approximation .
When the battery voltage is 6.5v I got 3.25v from the voltage divider and sample1 = 696 in serial monitor ,where sample1 is ADC value corresponds to 3.25v
For better understanding I have attached the real time simulation by 123D.circuit for voltage measurement
3.25v equivalent to 696
1 is equivalent to 3.25/696=4.669mv
Vout = (4.669*sample1)/1000 volt
Actual battery voltage = (2*Vout) volt
// taking 150 samples from voltage divider with a interval of 2sec and then average the samples data collected for(int i=0;i<150;i++)
sample1=sample1+analogRead(A2); //read the voltage from the divider circuit
Step 4: Current Measurement
For current measurement I used a Hall Effect current sensor ACS 712 (20 A).There are different current range ACS712 sensor available in the market, so choose according to your requirement. In bread board diagram I have shown LED as a load but the actual load is different.
WORKING PRINCIPLE :
The Hall Effect is the production of a voltage difference (the Hall voltage) across an electrical conductor, transverse to an electric current in the conductor and a magnetic field perpendicular to the current.
To know more about Hall Effect sensor click here
The data sheet of ACS 712 sensor is found here
From Data Sheet
1. ACS 712 measure positive and negative 20Amps, corresponding to the analog output 100mV/A
2. No test current through the output voltage is VCC / 2 =5v/2=2.5V
Analog read produces a value of 0-1023, equating to 0v to 5v
So Analog read 1 = (5/1024) V =4.89mv
Value = (4.89*Analog Read value)/1000 V
But as per data sheets offset is 2.5V (When current zero you will get 2.5V from the sensor’s output)
Actual value = (value-2.5) V
Current in amp =actual value*10
// taking 150 samples from sensors with a interval of 2sec and then average the samples data collected
sample2+=analogRead(A3); //read the current from sensor
actualval =val-2.5; // offset voltage is 2.5v
For more detail: Arduino Energy meter
This Post / Project can also be found using search terms:
- arduino energy meter code
- energy meter using arduino
- smart energy meter using arduino
- energy meter code