Teardown and Repair of an Agilent 33250A Function and Arbitrary Waveform Generator
In this episode Shahriar repairs an Agilent 33250A function and arbitrary waveform generator which does not power on. The unit is equipped with a soft power switch which might be responsible for the fault. After a brief look at the unit’s constructions, the fault is traced to a few possible locations.
The schematic of the soft power circuitry is presented and the internal switching power supply is closely examined. After the fault is located, the repair is presented. The block diagram and operation of a current mode DC-DC switching IC is also presented.